Operating Environment:Temperature: 0-40°C
Temperature Coefficent:0.05x(2%rdg±0.01mm,or 2%rdg±4mil)/°C (<18°C or >28°C)
Storage Environment:Temperature:-10°C to 50°C
1.Hold the tester and make sure that it is as least 20cm away from the object to be tested and other
objects.Then press the “TEST”button to turn on the tester.”–”appear on the
display and flashed,meanwhile the unit “mm”appear.
2.Press the “UNIT”button to select desired unit – mm or mil.
3.Place the tester’s probe perpendicularly on the surface to be tested,
the display will show the coating thickness.
*If the display shows”–”,it means that the tester does not detect any iron,steel or aluminum substrate.
*If the display shows”OL”,it means that the coating thickness exceeds the
tester’s measuring range.
*If the display shows”- 0″,it means that the probe has wear or the substrate under the coatings may be other
kind of metal other than iron,steel or aluminum.
4.When you make measurement,the reading will remain on the display if you
move qucikly the tester away from the surface under test;and you can press
the “TEST”button to delete this reading,the display will show”–”
5.If you do not press any button and the tester does not detect the presence of steel,
iron or aluminum for about 30 seconds,the tester will turn off automatically.
If you want to turn off the tester manually,press and hold down the “TEST” button
for more than about 1 second.